THE ANABAG 150™
|Cat. no. AB20||AnaBag 150™||20 bags/box|
Hardy Diagnostics AnaBag 150™ is a disposable environmental bag used for the incubation of 150mm plates in both anaerobic and CO2 enriched atmospheres. The AnaBag 150™ can be used as follows:
1. To hold a large (150mm) plate of HTM Agar (Cat. no. H07) for sensitivity testing for H. influenzae in a CO2 enriched atmosphere. This plate can hold up to 12 disks or 6 "Etest®" strips.
2. To hold a large (150mm) plate of Brucella with H & K (Cat. no. H05) for the "Etest®" strips for sensitivity testing of anaerobes in an anaerobic atmosphere. This plate can hold up to 6 "Etest®" strips.
The AnaBag 150™ is constructed from a triple laminated polymer which forms a barrier to oxygen, CO2, and other gases. This bag is not suitable for maintaining microaerophilic (5% O2, 10% CO2, 85% N2) conditions. The dimensions are 8 x 12 3/4 inches.
Store product in a dry, cool place away from extreme heat or freezing. Do not use if product is torn or damaged.
1. Inoculate plate and place disks or strips according to the manufacturers instructions.
2. Place the plate in the bag, then follow the procedure to activate gas generators, using the manufacturers instructions. Insert/Activate the color indicator as instructed by the manufacturer. See the "Materials Needed" section below.
3. The bag can then be sealed with a heat sealer or with a plastic sealing bar listed in the "Materials Needed" section below.
4. Incubate at an appropriate temperature and time for maximum recovery of the organism(s). The bag is opened by pulling the orange tab. Remove the plate and examine for growth.
The AnaBag 150™ should be discarded after each use, due to the possible development of pinhole size leaks or moisture leading to contamination.
INTERPRETATION OF RESULTS
A color change in the indicator strip, as well as growth of test organisms signifies that the proper atmospheric condition has been achieved. Consult the manufacturers package insert for additional information regarding interpretation of the indicator strip.
Indicator strips are designed to be used only once and then discarded. Once opened, indicator strips and gas generators should immediately be placed and sealed in the AnaBag 150™.
Inoculated media should be placed in the specified environment as soon as possible.
A tight seal must be made in order to generate and maintain the necessary atmospheric conditions.
This bag is not recommended for use in maintaining a microaerophilic (as for Campylobacter spp.) atmosphere.
MATERIALS REQUIRED BUT NOT PROVIDED
The following additional products are available from Hardy Diagnostics:
|1.||Culture Media||Diameter||Cat. no.||Organisms Tested|
|Brucella with H & K Agar||150mm||H05||All Anaerobes|
|Haemophilus Test Medium (HTM)||150mm||H07||Haemophilus spp.|
|Mueller Hinton with Blood||150mm||H21||S. pneumoniae|
|AnaeroGen™, Cat. no. AN25US and AN35US, is an anaerobic gas generating envelope that requires no water and no catalyst. It is packaged 10 envelopes/box.|
|CO2 Enriched Atmosphere|
|CO2 Gas Generators, Cat. no. 261205, 500/box, by Becton Dickinson.|
|CO2 Indicators - IndiKwik™, Cat. no. G511, 100 tests, by Pulse Scientific.|
|Cat. no. 260652, plastic, reusable, 10/box, by Becton Dickinson.|
USER QUALITY CONTROL
A color change in the indicator strip, as well as growth of test organisms signifies that the proper atmospheric condition has been achieved.
The AnaBag 150™ should be sealed on one end with an orange tab attached at the end of the seal.
AnaBag150™ (Cat. no. AB20).
|AB20||8 x 12 3/4 inch bag||20 bags/box||(gas generators, indicators, and sealing bars are not included)|
Etest is a registered trademark of AB BIODISK, a product distributed by:
AnaeroGen is a trademark of Oxoid, a product distributed by:
IndiKwik is a trademark of Pulse Scientific, a product distributed by: